Need of education for dry powder inhaler storage and retention – a patient-reported survey
© Norderud Lærum et al. 2016
Received: 4 March 2016
Accepted: 30 March 2016
Published: 8 June 2016
Dry powder inhalers (DPIs) are the most commonly used devices in asthma treatment in the Nordic countries. As new DPIs become available, patients are likely to be exposed to more than one type of device, with variable optimal handling. The aim was to examine real life storage and retention of multidose DPIs in patients with asthma.
This patient-reported survey on real life storage and retention of DPIs included asthma patients using multidose DPIs. Basic patient characteristics, information on inhaler use and storage, check of expiry date, and concurrent inhaler use was examined using an on line questionnaire.
A total of 738 patients were included with a median age of 41 years, out of which 83 % were women. Sixty-three per cent reported storage conditions pre-defined as risk locations for their maintenance inhaler and 38 % of the responding patients had more than one maintenance inhaler in use at the same time. Two thirds of the study population checked inhaler expiry date less than monthly or not at all. Use after expiry date was frequently reported. Two thirds of the patients had not received information on DPI storage, either from their doctor and/or nurse or at the pharmacy.
This patient reported survey indicates that two thirds of the patients store their inhaler devices in suboptimal conditions, and only a minority had received instruction regarding inhaler handling. Non awareness of inhalers’ expiry dates and use of more than one maintenance inhaler simultaneously was common. As inhaler mishandling may impact device functionality, improved communication and patient education is needed.
KeywordsReal life Survey Dry powder inhaler Asthma Control Storage Retention Compliance
A number of different devices are used for drug delivery to the lungs in the treatment of asthma, e.g. dry powder inhalers (DPIs), pressurized metered dose inhalers (pMDIs) and soft mist inhalers (SMIs). Of these, DPIs are the most commonly used inhalers in the Nordic countries. In Norway, the past years have witnessed an increase in available DPI devices to currently 12 different device providers. Combinations of inhaled corticosteroids (ICS) and long-acting beta2 agonists (LABA) were the top two reimbursed prescribed drugs in 2012 . Hence, treatment of asthma have a substantial impact on health care costs.
The asthma prevalence has increased in Norway over the last four decades, and is reported to be between 8 and 12 % [2, 3]. GINA guidelines recommend a stepwise approach to reach optimal asthma control . This should include not only receiving a correct diagnosis, but also a choice of inhaled therapy made in agreement between the physician and the patient, training on correct device use , and information on inhaler handling, storage, and retention. On the other hand, lack of treatment adherence and incorrect daily use may occur in up to 94 % of asthma patients [6, 7]. Hence, each of these steps are important to achieve asthma control.
Due to effective modern inhalation treatment, patients can today live a normal active life, and may have multiple inhalers available for use in different locations according to their lifestyle. Asthma patients are likely to be exposed to more than one type of inhaler concurrently during their treatment journey. Suboptimal inhaler technique  and inhaler mishandling  are still common in real life use. Lack of device continuity may increase these problems [10, 11]. To ensure optimal inhaler use, patient's preference  and education  and training of health care professionals in device handling  need to be in focus. The fact that inhalers differ in inhalation techniques, time in-use, and storage conditions needs increased awareness.
DPIs are breath actuated and more easy to use as there is no need to coordinate the inhalation, in contrast to most pMDIs. A potential drawback, however, is that DPIs can be sensitive to humidity, and in vitro studies have shown that the fine particle dose and the actual delivered dose may be negatively affected by storage in humid conditions [14, 15]. Differences in the resistance to humidity between different inhalers have been reported, likely due to the different technical approaches developed to protect the respective dry powder from humidity . A potential decrease of the dose reaching the lungs may eventually impact the clinical outcomes and infer decreased drug efficacy and safety. Still, the full clinical impact has not been indisputably established.
Number of doses, shelf life and storage of the most commonly used multidose dry powder inhalersa
Shelf life (unopened)
Time in-use (opened)
Recommended storage conditions
1.5–3 years (depending on drug strength)
Do not store above 30 °C. Store in a dry place.
Inhaler is sealed in a foil overwrap, which should only be opened when it is to be used for the first time.
Store in the original package.
When in use, do not store above 30 °C and store protected from moisture.
Do not store above 25 °C
Store in the original package in order to protect from moisture.
Keep the inhaler protected inside the sealed pouch until the administration period starts.
Do not store above 25 °C.
Store in the original package in order to protect from moisture.
Store in the original package.
When in use, keep the device tightly closed in order to protect from moisture.
Do not store above 25 °C.
Keep the mouthpiece cover closed after removal of the foil wrapping.
Do not store above 30 °C.
Study design and patient population
A patient reported survey was conducted in collaboration with a pharmacy chain (Boots A/S, Norway). No formal sample size calculation was performed. The pharmacy administered the survey via e-mail to all individuals who had indicated an interest in receiving information from the pharmacy on asthma and/or allergy. Based on the assumption that 50 % of the approximately 20,000 patients included in the pharmacy’s asthma and allergy customer register were asthmatics using DPIs, and with an expected response rate of 10 %, the sample size was estimated to 1,000 patients. Patients had to 1) confirm their diagnosis of asthma; and 2) accept participation in the study to have on line access to the self reported questionnaire. In addition, a patient-confirmed prescription of inhaled treatment by a multi dose DPI during at least 3 months prior to inclusion was required. The survey included questions on basic patient characteristics (such as age, gender, and number of concomitant inhalers for maintenance and as-needed use), and covered also different aspects of inhaler storage (e.g. actual storage conditions and instructions provided prior use) and retention (e.g. usage pattern and expiry date awareness). Primary maintenance defined as the most commonly used inhaler.
This patient-reported survey aimed at investigating the real life handling of multi dose dry powder inhalers for the treatment of asthma, including storage under non-optimal conditions, use after expiry date, use of concomitant inhalers, and how instructions regarding device handling were given.
The study variables included information such as time on DPI treatment, number of concurrent multi-dose DPI inhalers, frequency of inhaler expiry date checks, use of inhaler after expiry date, storage of inhaler in warm or humid conditions, and information on how patients were instructed in how to store and retain their inhaler devices. Data are presented using descriptive statistics.
Definition of storage conditions
The following storage conditions, predefined according to study protocol, were considered as risk locations for DPI storage: warm and humid conditions (summer outdoors, bathroom, gym, or indoor swimming pool), humid conditions (winter outdoors), and warm conditions (car, bag).
A total of 738 patients (median age 41 years; 83 % women) fulfilled the criteria and completed the self-reported questionnaire. Twenty-five percent (25 %) of the patients were between 36–45 years, 30 % were younger and 45 % of the respondents were older than 45 years. A total of 10 % were >65 years. The majority of patients (71 %) had used a DPI for more than 5 years, whereas only 4 % had used their DPI for between 3 months and 1 year. Overall, 81 % reported on having an inhaler for maintenance use. No patient characteristics or other information was available for the non-responders.
The primary maintenance inhaler was stored in the bathroom by 42 % of the patients and either in handbag or pocket by an additional 21 %. For the secondary maintenance inhaler the figures were almost the opposite, with 43 % storage in handbag or pocket, and 18 % in the bathroom. Overall, storage according to the predefined risk locations was reported by 63 % of the patients for the primary, 61 % for the second and 41 % for the third maintenance inhaler, respectively.
Patient-reported source of information on dry powder inhaler storage and retention, divided by gender and age below or equal to, or above 45 years
Thirty-eight per cent (38 %) of the responding patients stated that they had more than one maintenance inhaler in use at the same time. The majority of these (30 % of the study population) had 2 simultaneous maintenance inhalers, very few (2 %) used four or more inhalers concurrently. The majority of the patients (83 %) reported having a separate reliever medication for symptom control (as needed use), and one third of the patients (29 %) had more than one as needed inhaler in use at the same time. Most common storage place was in a bag, 42 %. Only 4 % of the patients reported storage of their as needed inhaler according to the instructions.
Available DPIs have differences in shelf life, time in-use and storage conditions. Failure to comply with the requirements may impact the functionality of the device. In this patient reported survey on asthma patients treated with multi dose dry powder inhalers, a substantial proportion of asthma patients have their DPIs stored in conditions with risk of exposure to humidity. The patients commonly use their DPIs after expiry date. Further, very few patients had received any information regarding inhaler storage and retention, and for the ones who had, the main source of information was the package insert, not their health care contact or pharmacist. Males were more likely to receive information at their health care center compared to females, which may reflect a gender-linked behavior in health care.
The present study comes with a number of limitations. The design is based on patient reported data only, hence asthma diagnosis and actual treatment with DPI could not be verified, and it is well known that self-reported medication use may be different than actual use. In internet-based research surveys, response representativeness is more important than response rate . In our study, the included population consists of individuals who had actively indicated interest in asthma and/or allergy and may represent an asthma population with more than average knowledge and motivation for asthma treatment. Hence, one could speculate that the knowledge of correct use and storage of inhalers in the general asthma population might be even lower than what is seen in our study. The questionnaire and the storage conditions were defined for the present study only, and were thus not validated or standardized. The gender distribution among the responders can be considered skewed. Although, the study has a low response rate, the results are well in accordance with other studies on real life challenges in asthma care.
Most inhalers have recommendations on storage temperatures and humidity, and some inhalers even need humidity protection storage until start of use (Table 1). Of the currently available DPIs, Diskus and Turbuhaler have the by far longest time in-use, but they may differ in tolerability to humidity. However, this study was not designed to evaluate quality differences between inhalers, but to investigate which conditions an inhaler is exposed to in real life and thus should be able to tolerate. The present study shows that 63 % of the responders store their maintenance inhaler(s) in sub-optimal storage conditions. This is well in keeping with a previous study, where required storage instructions for inhalation capsules was lacking by 84 % of the patients .
Concurrent use of multiple inhalers could potentially lead to prolonged use of the inhalers. Fewer inhalations from each inhaler results in a high number of doses left in the inhaler at the expiry date, encouraging continued use. In the present asthma study, a clear majority of the patients reported being more or less unaware of the expiry date of their DPI. This may be further complicated if the patient has more than one type of inhaler with different time in-use.
The present patient reported survey indicates that two third of the patients store their inhaler devices in suboptimal conditions, and only a minority had received instruction regarding inhaler handling. Two thirds of the study population checked inhaler expiry date less than monthly or not at all. Use of more than one maintenance inhaler simultaneously was common. Overall, this may impact the functionality of the device and result in sub-optimal treatment. Improved communication and patient education to ensure adherence to the different requirements of each inhaler device is needed.
This study was sponsored by AstraZeneca. Statisticon AB, an independent statistical consultant company for which AstraZeneca is a client, was responsible for the electronic case report form and the data analyses.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Folkehelseinstittutet. The Norwegian prescription database 2009–2013. ISBN: 978-82-8082-619-0 elektronisk utgave/electronic version (http://www.fhi.no/dokumenter/7e3ca9fffe.pdf). Accessed 18 Dec 2014.
- Brogger J, Bakke P, Eide GE, Johanssen B, Andersen A, Gulsvik A. Long-term changes in adult asthma prevalence. Eur Respir J. 2003;21(3):468–72.View ArticlePubMedGoogle Scholar
- Tollefsen E, Langhammer A, Romundstad PR, Bjermer L, Johnsen R, Holmen TL. Female gender is associated with higher incidence and more stable respiratory symptoms during adolescence. Respir Med. 2007;101(5):896–902.View ArticlePubMedGoogle Scholar
- GINA report, global strategy for asthma management and prevention. 2014; Available from: http://ginasthma.org. Accessed 19 Jan 2015.
- Grydeland TB, Methlie P, Bakke PS. Instructing patients in the correct use of inhalation devices (in Norwegian). Tidsskrift Nor Laegeforen. 2006;126(3):312–4.Google Scholar
- Lavorini F, Magnan A, Dubus JC, et al. Effect of incorrect use of dry powder inhalers on management of patients with asthma and COPD. Respir Med. 2008;102:593–604.View ArticlePubMedGoogle Scholar
- Mäkele MJ, Backer V, Hedegaard M, Larsson K. Adherence to inhaled therapies, health outcomes and costs in patients with asthma and COPD. Respir Med. 2013;107(10):1481–90.View ArticleGoogle Scholar
- Papi A, Haughney J, Virchow JC, Roche N, Palkonen S, Price D. Inhaler devices for asthma: a call for action in a neglected field. Eur Respir J. 2011;37:982–5.View ArticlePubMedGoogle Scholar
- Melani AS. Inhalatory therapy training: a priority challenge for the physician. Acta Biomed. 2007;78(3):233–45.PubMedGoogle Scholar
- Lavorini F, Ninane V, Haughney J, Bjermer L, Molimard M, Dekhuijzen RP. Switching from branded to generic inhaled medications: potential impact on asthma and COPD. Exp Opin Drug Deliv. 2013;10(12):1597–602.View ArticleGoogle Scholar
- Bjermer L. The importance of continuity in inhaler device choice for asthma and COPD. Respiration. 2014;88(4):346–52.View ArticlePubMedGoogle Scholar
- Dal Negro RW, Povero M. Acceptability and preference of three inhalation devices assessed by the handling questionnaire in asthma and COPD patients. Multidiscip Respir Med. 2016;11:7.View ArticlePubMedPubMed CentralGoogle Scholar
- Price D. The way forward: dry powder inhalers should only be switched with physician agreement and patient training. Int J Clin Pract Suppl. 2005;36–7.Google Scholar
- Asking L, Axelsson M, Lindberg J. Aluminium blisters may fail to protect against humidity. Drug Deliv Lungs. 1998;IX:84–7Google Scholar
- Borgström L, Asking L, Lipniunas P. An in vivo and in vitro study comparison of two powder inhalers following storage at hot/humid conditions. J Aerosol Med. 2005;18(3):304–10.View ArticlePubMedGoogle Scholar
- Cook C, Heath F, Russel, Thompson L. A meta-analysis of response rates in web- or internet-based surveys. Educ Psychol Meas. 2000;60:821–36.View ArticleGoogle Scholar
- van Renswouw DC, van Laarhoven ACM, van Haren MJ, Bouvy ML, Weda M. Storage instructions for inhalation capsules: consequences of incorrect storage and adherence in daily practice. J Pharm Pract. 2010;23(6):548.View ArticlePubMedGoogle Scholar